BeNano 90 Zeta
The BeNano Series is the latest generation of nanoparticle size and zeta potential analyzers designed by Bettersize Instruments. Dynamic light scattering (DLS), electrophoretic light scattering (ELS), and static light scattering (SLS) are integrated into the system to provide accurate measurements on particle size, zeta potential, and molecular weight. The BeNano Series is widely applied in academic and manufacturing processes of various fields including but not limited to: chemical engineering, pharmaceuticals, food and beverage, inks and pigments, and life science, etc.
Features and Benefits
 ● Size range: 0.3nm  15μm
 ● Minimum sample volume 3μL
 ● APD (Avalanche Photodiode) detector providing exceptional sensitivity
 ● Automatic adjustment of laser intensity
 ● Intelligent algorithm of result evaluation
 ● PALS (Phase Analysis Light Scattering) technology
 ● Programmable temperature control system
 ● Compliance with 21 CFR Part 11, ISO 22412, ISO 13099
Video
Bettersize BeNano Launch Event  Nanoparticle size and zeta potential analyzer
Product launch event  Nanoparticle analyzer on March 10th! BeNano Teaser
BeNano 90 Zeta  Nanoparticle size and zeta potential analyzer
Fundamentals of BeNano 90 Zeta
BeNano 90 Zeta  Demo (Polystyrene Standard Sample)
4 Questions Nanoparticle Researchers are Really Asking About
Overview
1) Optical Layout of the BeNano 90 Zeta
Dynamic light scattering (DLS), also known as photon correlation spectroscopy (PCS) or quasielastic light scattering (QELS), is a technology used to detect the fluctuations of the scattering intensities caused by the Brownian motion of particles.
In the dispersant, smaller particles move faster, while larger particles move slower. An avalanche photodiode (APD) detector aligned at 90°collects the scattering intensities of the particles and records them with time. The timedependent fluctuation is converted into a correlation function using the correlator. By applying a mathematic algorithm, the diffusion coefficient D is thereby obtained. The hydrodynamic diameter D_{H} and its distribution are calculated through the StokesEinstein equation.
3) Electrophoretic Light Scattering
Particles usually carry charges on the surface in aqueous systems, surrounded by counterions that form a firmly inner Stern layer and an outer shear layer. Zeta potential is the electrical potential at the interface of the shear layer. Suspension systems with higher zeta potentials tend to be more stable and less likely to form aggregates.
Electrophoretic light scattering (ELS) is a technology for measuring electrophoretic mobility via Doppler shifts of the scattered light. When an incident light illuminates dispersed particles that are subjected to an applied electric field, the frequency of the particles' scattered light will be different from the incident light due to the Doppler effect. The frequency shift is measured and converted to provide the electrophoretic mobility and hence the zeta potential of a sample by Henry's equation.
Static light scattering (SLS) is a technology that measures the scattering intensities of the sample, weightaverage molecular weight (Mw), and second virial coefficient A_{2} through the Rayleigh equation:
where c is the sample concentration, θ is the detection angle, R_{θ} is the Rayleigh ratio used to characterize the intensity ratio between the scattered light and the incident light at angle θ, Mw is the sample’s weightaverage molecular weight, A_{2} is the second virial coefficient, and K is a constant related to (dn/dc)^{2}.
During molecular weight measurements, scattering intensities of the sample at different concentrations are detected. By using the scattering intensity and Rayleigh ratio of a known standard (such as toluene), the Rayleigh ratios of samples at different concentrations are computed and plotted into a Debye plot. The molecular weight and the second virial coefficient are then obtained through the intercept and slope from the linear regression of the Debye plot.
5）Phase Analysis Light Scattering (PALS)
The traditional ELS converts the correlated scattering signals into frequency distribution and then calculates the frequency shift Δf of the scattered light, compared with the reference light. Phase analysis light scattering (PALS), an advanced technology based on the traditional ELS technology, has been further developed by Bettersize Instruments Ltd. to measure zeta potential and its distribution of a sample. By analyzing the phase information Φ of the original scattered signal, PALS obtains the frequency information of that light. The phase shift with time dΦ/dt is proportional to the frequency shift Δf. PALS technology can suppress the influence of the Brownian motion of particles on the results, thereby providing higher statistical accuracy. In various applications, PALS can effectively measure the zeta potential of particles whose charge approaches the isoelectric point, for instance, particles with very slow electrophoretic mobility at a high salt concentration.
6) Temperature Trend Measurement
Temperature trend measurement includes:
 Size vs. Temperature
 Zeta Potential vs. Temperature
Investigating the particle size and zeta potential of the samples under different temperatures is significant in many applications. The function of Programmed Temperature Change, ranging from 10°C to 110°C, makes Temperature Trend Measurement available in the BeNano Series.
Benefits
 This feature benefits users who need to study the stability of protein formulations. Generally, the higher the denaturation temperature of the protein, the more stable the formulation.
 Besides, it is useful for users who need to simulate realtime aging using elevated temperatures to manually speed up the aging process.
7) Viscosity Measurement
For the sample of unknown viscosity, the viscosity measurement could be implemented using tracer particles with known sizes (e.g., standard samples with nominal sizes). When the measurement ends, input the accurate size of the tracer particles, and the viscosity of the sample could be determined.
After the measurement, choose and rightclick on the corresponding result. Click "Viscosity Calculator" on the popup menu. By inputting the nominal values of the tracer particles and clicking on "Calculation", the viscosity of the sample could be finally ascertained.
Curated Resources
 Application NotebigClick
20230602
Using the BAT1 autotitrator to measure the zeta potentials of TiO2 at different pH
 GuidebookbigClick
20230309
Sample preparation guide for wet measurement using the laser diffraction particle size analyzer
 GuidebookbigClick
20230116
A Practical Guide to Nanoparticle Characterization by Light Scattering Techniques
Related Nanoparticle Size and Zeta Potential Analyzer

BeNano 180 Zeta Pro
Nanoparticle Size and Zeta Potential Analyzer
Technology: Dynamic Light Scattering, Electrophoretic Light Scattering, Static Light Scattering